Back to results list
Please use this identifier to cite or link to this item:
|Title:||Validation of keratometric measurements obtained with a new integrated aberrometry-topography system||Authors:||Shneor, E
|Issue Date:||2012||Source:||Journal of optometry, 2012, v. 5, no. 2, p. 80-86||Abstract:||Purpose: A clinical evaluation of the L80 videokeratographer (Visionix Luneau, Chartres, France) was performed to assess its validity and repeatability compared with a traditional Bausch and Lomb (B & L) keratometer. Methods: 87 right eyes of 87 subjects, (mean age 23.72 ± 3.62 years old, 70 women and 17 men), participated in this study. Corneal curvature was measured using the L80 instrument by one practitioner and the manual B & L keratometer by a different practitioner. Intratest and intertest repeatability were assessed. Results: Corneal curvature was found to be statistically different between the two instruments (p < 0.001), with the L80 providing a slightly steeper bias of 0.05 mm and 0.07 mm for the horizontal and vertical meridians, respectively than the B & L keratometer. 78.2% and 86.2% of the L80 results were within ±0.1 mm (±0.06 D) and 95.4% and 97.7% within ±0.2 mm (±0.11 D) of the readings obtained with the B & L keratometer along the horizontal and the vertical meridians, respectively. The agreement between the L80 and B & L keratometers axes was 31.0% within ±5°, 54.0% within ±10°, 60.9% within ±15°, 71.3% within ±20° and 87.4% within ±40°. Intratest repeatability was the same for both instruments. Intertest repeatability was better for the L80 videokeratographer compared to the B & L keratometer and showed no significant difference between the two sessions. Conclusion: The L80 videokeratographer is a reliable objective instrument comparable to other autokeratometers which, in addition, combines many other useful clinical features. It provides steeper radii of curvature measurements than the B & L keratometer. An offset incorporated into the instrument could mitigate the difference between the two instruments and make them interchangeable.||Keywords:||Keratometry
Limits of agreement
|Publisher:||Elsevier Doyma||Journal:||Journal of Optometry||ISSN:||1888-4296||DOI:||10.1016/j.optom.2012.03.003||Rights:||© 2012 Spanish General Council of Optometry. Published by Elsevier España, S.L.
This is an open access article under the CC BY-NC-ND license (http://creativecommons.org/licenses/by-nc-nd/4.0/)
The following publication Shneor, E., Millodot, M., Zyroff, M., & Gordon-Shaag, A. (2012). Validation of keratometric measurements obtained with a new integrated aberrometry-topography system. Journal of Optometry, 5(2), 80-86 is available at https://doi.org/10.1016/j.optom.2012.03.003
|Appears in Collections:||Journal/Magazine Article|
Show full item record
Files in This Item:
|Shneor_Validation_Keratometric_Measurements.pdf||350.37 kB||Adobe PDF||View/Open|
Citations as of Jun 4, 2023
Citations as of Jun 4, 2023
Citations as of Jun 8, 2023
Items in DSpace are protected by copyright, with all rights reserved, unless otherwise indicated.